The hot-filament ionization gauge is widely used for measuring the level of vacuum. The electrons are emitted from a hot cathode and accelerated towards the anode. In that process the electron may ionize a gas molecule. That gas molecule is pulled towards the collector and ion current is measured with an ammeter.
Vacuum can be understood as space from where matter (for example air) has been removed. It naturally exists in outer space but for certain applications, like materials characterization techniques, it needs to be achieved artificially. The desired level of vacuum is obtained with the help of a suitable vacuum pump. For example low vacuum (low quality vacuum with higher pressure) can be generated with a diffusion pump, scroll compressor pump, rotary vane pump, diaphragm pump or a sorption pump. High vacuum (high quality vacuum with very low pressures) however, can be obtained with high vacuum pumps such as the turbomolecular pump, ion pump, titanium sublimation pump and cryopump. The level of vacuum is measured with devices called vacuum gauges (vacuum meters) like the thermocouple gauge, pirani gauge, penning ionization gauge and the quadrupole mass spectrometer (analyzer). The working principle of vacuum pumps and vacuum gauges is explained with 3D animations in the video lecture above.